Question in Detail:
I just received a Baader 2" Clicklock TAK-adapter M72. After a few uses it's spinning where it shouldn't. If I hold the inside on the bottom near where it threads onto the scope it works. Can you tell me what to do? Can I repair it?
This might be a mistake done here during assembly. Please try this fix. If it does not help please return the Clicklock clamp to us or to your dealer, we apologizie.
Actually the 2"/M72 ClickLock is consisting of the M68 ClickLock and the Adapter M68/M72. There are 6x M2.5 screws underneath the M68/M72 Adapter. These prevent the M68 thread from rotating. It seems that they where not properly tightened. If you grab the very small outer ring on the ClickLock which is part of the M68 Adapter, and remove the M72 Adapter, then you can fasten the M2.5 screws. We hope this helps.